KEITHLEY 2182A NANOVOLTMETER LABVIEW DRIVER
It is possible that there is some setting that needs to be made on these particular devices to voltage sweep a resistor versus a transistor. It is an example software written by the Keithley. Unfortunaltely keithley engineers are strongly committed not to provide the block diagram. Can you tell me exactly what is “auto-test sequence running? Message 9 of
|Date Added:||13 September 2017|
|File Size:||56.78 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Please attach this VI to the reply of this post or point me in the direction to where you had found it. Can you tell me exactly what is “auto-test sequence running? Nudge I like Keithley intsruments but they are going to obsolete themselves if they don’t snap to it and write drivers that work.
It is an example software written by the Keithley. I feel people have written some program like KIx.
Keithley 2182 Meter
The Keithley driver is not a supported National Instruments driver, but I’ll try to help you the best I can. I tried nanovolmteter the program in “Demo mode” with your same settings, and the run was successful, without throwing a warning.
But I could not figure out how to set it right. This voltage sweep function works nanovoltmetre over a resistor, but not over a transistor. I am trying to use the “differential conductance mode” in software “KI Does anyone has the block diagram for KIx.
Thanks for your reply.
keithley & A sweep in terms of voltage – Discussion Forums – National Instruments
Unfortunaltely keithley engineers are strongly committed not to provide the block diagram. Appreciates for any advices. Most Active Software Boards: It is possible that there is some setting that needs to be made on these particular devices to voltage sweep a resistor versus a transistor.
I think that the plus keifhley the conductance measurement software from Keithley are designed for resistor-like device: Message 7 of Message 9 of Message 10 of There is no manual of KI You can contact an engineer at Keithley here. Message 8 of This may be the source of the warning; the sweep is starting and stopping laview 0 A.
Is this a VI that you had created? In my case, I need to plot a 3-D graph which has source-drain voltage Xgate voltage Y and conductance Z.
Keithley Nanovoltmeter – Instrument Driver – National Instruments
The problem happened when I chose the sweep in terms of voltage rather than current. Since this is an executable and I am unable to view the Block Diagram, I cannot troubleshoot the code. Message 1 of Message 5 of Message 4 of I want to edit the program and make it convenient to me, as I suppose with my limited labview knowledge it will take more time if I strat from scratch.
Message 2 of I noticed you selected 0 A for both the current start and stop values. For some reason, I was unable to open the data sheets and manuals on Keithley’s website.
At this point, you may want to contact Keithley. I don’t know if you can setup such a “transistor”measurement!?